 PRODUCTS
Large Displacement Range
Jseries
Jseries accommodates high velocity and large displacement testing
Long duration shock tests require high velocity and large displacement.
Jseries is a highfunctionality system that offers usability and durability furnished with functions that accommodate high velocity and large displacement testing.
Features
 ■Expanded maximum test range
 ・Maximum velocity of SINE force:94 in/s ・Maximum velocity of SHOCK force:181 in/s ・Maximum displacement:4.0inpp
 ■Patented upper (armature) support system PS Guide
 Parallel Slope Guide is standard
 ■Low noise
 Optimised design of the air intake based on fluid dynamics has reduced the airintake noise.
 ■All models can be directly coupled to a climatic test chamber
Table Insert Pattern (Unit:mm)
Specifications
System Model  J230/ SA3M 
J230S/ SA7M 
J240/ SA4M 
J240S/ SA9M 
J250/ SA5M 
J250/ SA6M 
J260/ SA7M 
J260S/ SA30M 


Systme Specifications 
Frequency Range(Hz)  03000  03000  02400  02400  02200  02200  02600 ^{*4}  02000  
Rated Force 
Sine(lbf)  3600  3600  5400  5400  7900  9000  12150  12150  
Random (lbf rms) ^{*1} 
3600  3600  5400  5400  7900  9000  12150  12150  
Shock(lbf)  9000  9000  12400  15700  15800  18000  24200  44100  
Maximum Acc. 
Sine(g)  96  90  94  87  79  90  87  87  
Random (g rms) 
67  63  65  61  55  63  61  61  
Shock (g peak) 
240  226  216  254  159  181  174  255  
Maximum Vel. 
Sine(in/s)  94  94  94  94  94  94  94  94  
Shock (in/s peak) 
94  138  94  142  94  94  94  181  
Maximum Dsp. 
Sine(inpp)  4.0  4.0  4.0  4.0  4.0  4.0  4.0  4.0  
Maximum Travel(inpp) 
4.7  4.7  4.7  4.7  4.7  4.7  4.6  4.6  
Maximum Load(lbs)  660  660  880  880  1320  1320  2200  2200  
Power Requirements (kVA) ^{*2} 
28  38  38  52  53  57  86  127  
Breaker Capacity (A) ^{*3} 
100  150  150  200  200  200  300  500  
Vibration Generator 
Model  J230  J230S  J240  J240S  J250  J250  J260  J260S  
Armature Mass(lbs)  37.5  39.7  57.3  61.7  99.2  99.2  139  139  
Armature Diameter(φin)  7.9  7.9  11.4  11.4  17.3  17.3  17.6  17.6  
Allowable Eccentric Moment(lbf･in) 
6200  6200  7530  7530  13730  13730  13730  13730  
Dimensions(in) WxHxD 
45x 43x 34 
45x 43x 34 
49x 45x 35 
49x 45x 35 
58x 51x 44 
58x 51x 44 
60x 52x 44 
65x 52x 44 

Shaker Body Diameter (φin) 
25  25  29  29  34  34  36  36  
Mass(lbs)  3970  3970  5295  5291  7720  7720  9040  11025  
Power Amplifier 
Model  SA3M J 30 
SA7M J 30S 
SA4M J 40 
SA9M J 40S 
SA5M J 50 
SA6M J 50 
SA7M J 60  SA30M J 60S  
Maximum Output(kVA)  23  30  34  40  50  57  70  76  
Dimensions(in)WxHxD  23x 69x 33 
23x 77x 34 
23x 69x 34 
46x 77x 34 
23x 77x 34 
23x 77x 34 
23x 77x 34 
91x 77x 34 

Mass(lbs)  730  1105  970  2650  1940  2010  2205  7055  
Controller  Vibration Controller  See Vibration Controller K2  
Cooling  Cooling Method  Air cooling  
Blower  Dimensions(in) WxHxD^{*6} 
24x 52x 36 
24x 52x 36 
28x 61x 37 
28x 61x 37 
42x 73x 45 
42x 73x 45 
53x 85x 43 
53x 85x 43 

Mass(lbs)  371  371  474  474  827  827  898  898  
Wattage(kw)  3.7  3.7  5.5  5.5  11  11  15  15  
Duct Hose Diameter(φ) 
200  200  200  200  250  250  250  250 
Eco Specifications
System Model  EM2305  EM2405  EM2505  EM2506  EM2605  

System Specifications 
Frequency Range(Hz)  03000  02400  02200  02200  02600 ^{*4}  
Rated Force 
Sine(lbf)  3600  5400  7900  9000  12150  
Random (lbf rms) ^{*1} 
3600  5400  7900  9000  12150  
Shock(lbf)  9000 
12400 
15800 
18000 
24200 

High Velocity Shock (kN) 
6750  10800  15300  17520  21595  
Maximum Acc. 
Sine(g)  96  94  79  90  87  
Random (g rms) 
67  66  55  63  61  
Shock (g peak) 
240  216  159  181  174  
High Velocity Shock (g peak) 
180  188  154  176  155  
Maximum Vel. 
Sine(in/s)  94  94  94  94  94  
Shock (in/s peak) 
94  94  94  94  94  
High Velocity Shock (in/s peak) 
137  137  137  137  137  
Maximum Dsp. 
Sine(inpp)  4.0  4.0  4.0  4.0  4.0  
Maximum Travel(inpp) 
4.7  4.7  4.7  4.7  4.6  
Maximum Load(lbs)  660  880  1320  1320  2200  
Power Requirements(kVA) ^{*2}  28  38  53  57  86  
Breaker Capacity(A) ^{*3}  100  150  200  200  300  
Vibration Generator 
Model  J230  J240  J250  J250  J260  
Armature Mass(lbs)  37.5  57.3  99.2  99.2  139  
Armature Diameter(φin)  7.9  11.4  17.3  17.3  17.6  
Allowable Eccentric Moment(lbf･in) 
6200  7530  13730  13730  13730  
Dimensions(in)WxHxD  45x 43x 34 
49x 45x 35 
58x 51x 44 
58x 51x 44 
60x 52x 44 

Shaker Body Diameter (φin) 
25  29  34  34  36  
Mass(lbs)  3970  5295  7720  7720  9040  
Power Amplifier 
Model  SA3M J30EM 
SA4M J40EM 
SA5M J50EM 
SA6M J50EM 
SA7M J60EM 

Maximum Output(kVA)  23  34  50  57  70  
Dimensions(in)WxHxD  23x 69x 34 
23x 69x 34 
23x 83x 34 
23x 83x 34 
46x 77x 34 

Mass(lbs)  840  1080  2050  2120  3090  
Controller  Vibration Controller  See Vibration Controller K2  
Cooling  Cooling Method  Air cooling  
Blower  Dimensions(in) WxHxD^{*5} 
28x 56x 31 
28x 61x 37 
51x 84x 33 
51x 84x 33 
52x 89x 32 

Mass(lbs)  309  474  644  644  673  
Wattage(kw)  3.7  5.5  11  11  15  
Duct Hose Diameter(φ) 
200  200  250  250  250 
 *1 Random force ratings are specified in accordance with ISO5344 conditions. Please contact IMV or your local distributor with specific test requirements.
 *2 Power supply: 3phase 200/220/380/400/415V, 50/60 Hz. A transformer is required for other supply voltages.
 *3 In case of 200 V
 *4 Above 2000 Hz, the force rollsof f at a rate of 12 dB/oct.
 *5 Spec described above is in case of 60 Hz. Dimension is changed in case of 50 Hz.
 * The specification shows the maximum system performance. For longduration tests, derating by up to 70 % must be applied. Continuous use at maximum levels may cause failure.
 * In the case of Random vibration test, please set the test definition of the peak value of acceleration waveform to be operated less than the maximum acceleration of Shock.
 * Frequency range values vary according to sensor and vibration controller.
 * Mass and dimensions may change for CEmarked systems.